Application Note: Moisture Determination of Liquid Soap
Maintaining optimal moisture levels during the production of soap is critical to achieving the highest quality product. When the moisture content becomes too high or too low, product quality becomes inferior and waste occurs. For accurate moisture determination and regulation in your product, you need a top quality soap moisture analyser.
This information is intended as a guideline for development of a moisture analysis and content determination method and parameter setting. Materials vary throughout the world and we will be more than happy to test customers’ individual samples.
Precisa Moisture Determination XM60/EM120 Halogen
*Can also be conducted using XM50, XM60-HR & XM66
|Description:||Liquid soap, creamy liquid|
|Utensils:||Spatula, Glass fibre filter, Aluminium pan|
|Preparation:||Spread thinly and evenly onto a glass fibre filter|
|Sample weight (+/-10%):||2g|
|Stop-Criteria:||Autostop 2 Digits [mg] / 30 s or 0.1 %/30s|
(average of 6 measurements)
|Standard Deviation (+/-):||0.07%|
|Average analysis time:||7.4 min|
Download the Application Note
You can download the complete application note as a pdf: Moisture Liquid Soap Application Note
Precisa Moisture Analyser XM60/EM120
To find out more about our Series 330 XM or our Series 365EM for moisture content determination methods, then please don’t hesitate to contact our sales and marketing team. They’ll be pleased to discuss your requirements.